- Blades, scalpels, stitch cutter (10)
- Laboratory Consumables (49)
- Blood Bags/Transfusion (9)
- Reagent/Dropper Bottle (11)
- Diagnostic Instruments (14)
- Examination Gloves (9)
- Hospital Equipment (5)
- Kit series/First Aid Kit (25)
- Medical Consumables (44)
- medical non-woven product (28)
- Medical Wound Dressing (22)
- Other Medical Products (5)
- Syringe/needles (16)
- Surgical Instruments (7)
- obstetrical instruments (4)
- Thermometer (3)
- Syringe Needle Destroyer (2)
- SPIROMETER (1)
- blood pressure monitor (4)
No. of Employees : 200~500 86-1860-6940498
hot-selling Medical oxygen regulator
|Place of Origin:||china|
|Payment Terms:||western union/TT/LC|
|Minimum Order Quantity:||1000|
|Packaging Details:||depend on customs requirements|
|Delivery Time:||depend on order quantity|
|Minimum Order Quantity:||50|
- Product Detials:
- 1. Buoy type oxygen inhalator: XY-98B, XY-98BIII Regulation range of oxygen flow 1-10L/min high pressure part can support 15MPa when the flow is shut off.
- Pressure can be controlled between 0.2-0.3MPa. XY-98BII, Input pressure: 0.35MPa Input Flow: 25L/min Flow regulation range of inhaling device: 0-15L/min Negative device suction pressure regulation range 0-350mbar
- 2. Aluminum Alloy Oxygen Regulator 1010 type
- Input: 2200PSI Output: 50PSI
- Flux: 12 levels in adjustment MINI 1010 type 22PSI
- Input: 2200PSI
- Output: 22PSI Flux: 12 levels in adjustment 1030 type,
- Input: 2200PSI
- Output: 50PSI
- Flux: 12 levels in adjustment There is British Connection, Z connection, etc
How to use
1. Open the valve of oxygen cylinder after connect the screw nut of air intake closely.
2. Connect the oxygen pipe to the oxygen outlet firmly.
3. Demount the humidifying cup(anti clock wise rotation), put some distilled water in (purified water or plain boiled water can also take place), the water quantity should be between the bottom and top tick mark, then tighten up the bottle cap to the humidifying cup(clockwise rotation).
4. Rotate the flow regulating valve to the rate you need (anti clockwise to up, clockwise to down, till to close), then put it to nose to use.
5. Close the valve of oxygen cylinder firstly while not using, then close the flow regulating valve until the pointer of oxygen manometer return to zero.
1. Make sure there is no ash on the connector and check the screw thread, then install the inhalator.
2. Make sure the flow meter is vertical to the floor and flow regulating valve is closed, then open the valve of oxygen cylinder slowly, regulate the flow after the high pressure gauge shows the pressure of the oxygen cylinder.
3. No drastic shock, far away with the corrosive substance, and no oil stain.
4. The set screw of the safety valve can't be rotated at random, non-professional people are forbidden to dismount.
Condition for storage
Should be kept in the environment whose humidity is less than 90%, no corrosive gas and drafty.
Quality guarantee period
We will fix or change the oxygen inhalator within a half and a year after the day of living the company when it can't proper functioning, it should be under the condition of obeying the rule of storage and using.
1. Forbid to use the Buoy Type Oxygen Regulator to take oxygen atomization treatment.
2. Forbid to dismount by yourself.
3. The defects liability period is 18 months.